Professional 3D Hifu Anti-Aging Facial Beauty Machine for Sale
|Mode||High Intensity Focused Ultrasound|
|Display||15.0 Inches Touch Screen|
|Voltage||AC110 - 240V, 50 / 60Hz|
|Hifu Energy||0.2-2.0j(0.1j/Step) Adjustable|
|Standard Hifu Cartridge||3.0mm / 4.5mm|
|Optional Hifu Cartridge||1.5mm / 6.0m/8.0mm/10mm/13mm/16mm|
|Hifu Shot Line||1-11 Lines|
|Hifu Width||10mm Max Adjustable|
|Transport Package||Standard Aluminum Box|
|Production Capacity||500 Sets Per Month|
1. After using the instrument , please clean it with alcohol disinfection and keep it properly .
2. The plug with grounding pin must be used before use , and ensure that the power socket of the instrument is truly grounded .
3. Please make sure that the voltage of instrument is adapted . If the local power supply voltage is unstable , we recommend that user increase the power matching power supply .voltage is unstable , we recommend that the user increase the power matching power supply .
4. In order to ensure the therapeutic effect and normal service life of the instrument , please use the specified accessories provided or recommended by the original manufacturer .
5. Do not place in a humid place or near a water source , and do not expose the instrument directly to the sun .
6. Do not place the instrument close to a strong heat source as this may affect the life of the insrument and its normal use .
7.Please first remove all metal objects from your body to avoid unprredictable conditions and affect the efficacy .
8.In order to avoid contact failure , please pay attention to the skin contact with a certain degree of moisture , with water based products or water to wipe it can achieve the desired effect .
9.When not in use , please turn off the power switch of the instrument , and ensure that the total power of the instrument is turned off after use to ensure the safety of the electrical peducts .
10.Using instruments or training instrument operators in strict accordance with instrutions in the use manual .